Class I Recall of CareFusion Alaris PC Unit, Model 8015
According to the recall notice a component on the PC unit power supply board is causing an error code (120.4630): "System Error" or "Missing Battery Error" to occur. The error code is accompanied by both an audible alarm and a visual error message on the PC unit screen. If the error code occurs at start-up, the Alaris infusion pump cannot be programmed and this may cause a delay in patient therapy. If the error code occurs during infusion, the health care provider cannot make programming changes to current infusions, and this may result in serious injury and/or death to the patient.
Want to read the full alert and receive alert emails?